Many surplus scintillation probes have a single connector through which the PMT is fed with high voltage and the anode signal is output. However, this may require an external “Bias-T” (a high voltage / signal splitter) to connect the probe to a high-voltage power supply that is separate from the PMT amplifier/processor. Continue reading
These two images supplement the book’s Figure 81. They were taken with the spectrometer of Figure 80.