Ludlum general-purpose ratemeters are professional-grade instruments that are available on the secondary market at affordable prices. They are compatible with a wide variety of probes, making them a great choice for educators, surveyors, and advanced amateur users. However, probes for Ludlum ratemeters are often as expensive as the meter instrument itself, making it worthwhile
We just finished constructing a low-cost, yet highly sensitive gamma-ray scintillation probe for our CDV700-Pro counter. The probe is based on a Philips XP5312/SN photomultiplier tube (that is available from Sphere Research) and a piece of scintillation plastic. The probe yields a background count of approximately 1,000 counts/minute (cpm) in our lab, and 7,400 cpm
Images Scientific Corporation just announced a new Geiger counter wand base provides a stable platform to hold the Geiger counter wand for experiments. The physical dimensions of the holder are 2″ wide by 12″ long. The length of the sled has markings in both metric and imperial. The sled isotope holder provides a stable and moveable
I purchased two “Giant Super Sensitive MC6” GM tubes from Electronic Goldmine (Item Number : G18717, Unit Price: $89.95). These are Russian-made new-old-stock model MC6. They are 10.25″ long x 0.9″ diameter. I compared the sensitivity of these tubes to the other GM tubes that I use with my CDV700 Pro Geiger Counter.
The book’s Figure 65 shows our β-particle magnetic deflection setup. It consists of a 90Sr disc source of beta particles, two copper washers to collimate the beam, and GM tubes placed at 0º and 90º to the β-particle beam. A sufficiently strong magnetic field (around 800 Gauss = 0.08 Tesla) provided by a permanent magnet bends the beam
The attenuation of radiation as a function of distance can be measured using a radiation counter with a Geiger-Müller tube that is sensitive to α, β, and γ radiation. We used exempt plastic-disc sources containing Polonium 210 (210Po), Strontium 90 (90Sr), and Cobalt 60 (60Co) to experiment with the penetrating power of α, β,
We modified a surplus Civil Defense V-700 radiation survey meter made by Electro Neutronics Inc. (Model 6-b) into a very capable radiation counter capable of working with both Geiger-Müller and PMT scintillation probes. We modified the front panel to accommodate the new switches, connectors, and panel light. In addition, we placed a Veeder-Root count totalizer